Reprinted from the December 1995 issue of RESPIRATORY CARE [Respir Care 1995;40(12):1336–1343]
AARC Clinical Practice Guideline
This Guideline was developed jointly by the AARC Cardiopulmonary Diagnostics CPG Focus Group and representatives of the Association of Polysomnography Technologists (APT). Both groups have approved its content.
PSG 1.0 PROCEDURE:
PSG 2.0 DESCRIPTION/DEFINITION:
Polysomnography refers to the collective process of monitoring and recording physiologic data during sleep. The specific variables monitored during center-based polysomnographic evaluation of sleep-related respiratory disturbances are listed.(1,2) Home-based polysomnography(3) and unattended monitoring systems are not addressed in this guideline.
The variables monitored and recorded during polysomnography include but are not limited to:(4,5)
- 2.1 global neural electroencephalographic activity (EEG) from electrodes placed on the patient's scalp;
- 2.2 eye movements (electro-oculogram, or EOG) from electrodes placed near the outer canthus of each eye;
- 2.3 submental electromyographic activity (EMG) from electrodes placed over the mentalis, submentalis muscle, and/or masseter regions;
- 2.4 rhythm electrocardiogram (ECG) with two or three chest leads;
- 2.5 respiratory effort, by chest-wall and abdominal movement via strain gauges, piezoelectric belts, inductive plethysmography, impedance or inductance pneumography, endoesophageal pressure, or by intercostal EMG;
- 2.6 nasal and/or oral airflow via thermistor or pneumotachograph;
- 2.7 oxygen saturation (SpO2) via pulse oximetry;
- 2.8body position via mercury switches or by direct observation;
- 2.9limb movements (arms and legs) via EMG;(6)
- 2.10 recordings of or vibration (frequency and/ or volume) may be recorded;(7)
- 2.11 end-tidal CO2, transcutaneous CO2, esophageal pH, penile tumescence, and bipolar EEG are beyond the scope and intent of this guideline.
PSG 3.0 SETTING:
Center-based polysomnography is performed within specialized hospital sleep laboratories,(1) appropriately equipped hospital rooms, or stand-alone sleep centers, with a qualified technician in constant attendance.(8)
PSG 4.0 INDICATIONS:
Polysomnography may be indicated in patients(9-12)
- 4.1with COPD whose awake PaO2 is > 55 torr but whose illness is complicated by pulmonary hypertension, right heart failure, polycythemia, or excessive daytime sleepiness;
- 4.2with restrictive ventilatory impairment secondary to chest-wall and neuromuscular disturbances whose illness is complicated by chronic hypoventilation, polycythemia, pulmonary hypertension, disturbed sleep, morning headaches, or daytime somnolence and fatigue;
- 4.3with disturbances in respiratory control whose awake PaCO2 is > 45 torr or whose illness is complicated by pulmonary hypertension, polycythemia, disturbed sleep, morning headaches, or daytime somnolence and fatigue;
- 4.4with nocturnal cyclic brady- or tachyarrhythmias, nocturnal abnormalities of atrioventricular conduction, or ventricular ectopy that appear to increase in frequency during sleep;
- 4.5with excessive daytime sleepiness or insomnia;
- 4.6with snoring associated with observed apneas and/or excessive daytime sleepiness;
- 4.7with other symptoms of sleep-disordered breathing as described in International Classification of Sleep Disorders;(6)
- 4.8with symptoms of sleep disorders described in the International Classification of Sleep Disorders.(6)
PSG 5.0 CONTRAINDICATIONS:
There are no absolute contraindications to polysomnography when indications are clearly established. However, risk-benefit ratios should be assessed if medically unstable inpatients are to be transferred from the clinical setting to a sleep laboratory for overnight polysomnography.
PSG 6.0 PRECAUTIONS/COMPLICATIONS:
- 6.1 Skin irritation may occur as a result of the adhesive used to attach electrodes to the patient.
- 6.2 At the conclusion of the study, adhesive remover is used to dissolve adhesive on the patient's skin. Adhesive removers (eg, acetone) should only be used in well-ventilated areas.
- 6.3 The integrity of polysomnographic equipment's electrical isolation must be certified by engineering or biomedical personnel qualified to make such assessment.
- 6.4 The adhesive used to attach EEG electrodes, (eg, collodion) should not be used to attach electrodes near the patient's eyes and should always be used in well-ventilated areas.
- 6.5 Due to the high flammability of collodion and acetone, they should be used with caution, especially in those patients who require supplemental oxygen.
- 6.6 Collodion should be used with caution in those patients with reactive airways disease and in small infants.
- 6.7 Patients with parasomnias or seizures may be at risk for injury related to movements during sleep. Institution-specific policies and guidelines describing personnel responsibilities and appropriate responses should be developed.
PSG 7.0 DEVICE LIMITATIONS/VALIDATION OF RESULTS:
Due to the nature of the various physiologic variables that are examined during polysomnography, a variety of devices are attached to the patient, and the output of these devices is interfaced with a standard polysomnographic recorder; consequently, the potential exists for several individual device limitations, rather than any one specific limitation to polysomnography.
- 7.1 Records of EEG, EOG, and EMG activity. These variables are monitored by gluing or taping, small metal-disk electrodes to the patient.(13) The low impedance limit is 1,000 ohms (1 K), and, in the routine diagnostic polysomnogram recorded at 10-15 mm/s using referential EEG channels, an impedance < or = 10,000 ohms (10 K) is acceptable for all channels.(14-16) Slightly higher impedances are acceptable for limb EMG.(17) However, a maximum impedance of 5,000 ohms (5 K) is recommended for bipolar EEG channels frequently used in extended polysomnography montages when seizure activity is suspected.(18) Limitations associated with EEG, EOG, and EMG monitoring include:
- 7.1.1 displacement of the electrode from the subject's skin causes a reduction, or complete loss, of the signal;
- 7.1.2 failure to properly prepare the skin or drying of the transduction gel inside of the electrode obscures the signal's characteristics;
- 7.1.3 artifactual signals can obscure actual physiologic signals;
- 7.1.4 environmental electrical noise can obscure actual physiologic signals.
- 7.2 Record of ECG activity. Rhythm ECGs are typically obtained by using two of the described EEG electrodes. Recording a modified Lead-II (right shoulder and left leg) or precordial lead (MCL) is sufficient. Limitations associated with rhythm ECGs obtained during polysomnography include:
- 7.2.1 The recording paper speed used during polysomnography may be slower than that used by cardiac monitors and recorders. This or improper filtering techniques may obscure P-wave or QRS-complex wave morphology.(5)
- 7.2.2 Other limitations can be the same as described in Section 7.1.
- 7.3 Record of chest-wall and abdominal movement. A variety of devices exist for measuring chest-wall and abdominal movement including mercury strain gauges, piezoelectric belts, inductance, and impedance pneumography.(1)
- 7.3.1 Mercury strain gauges are placed on or around the chest or abdomen. Limitations of strain gauges include:(19)
- 184.108.40.206 The quality and interpretability of the respiratory signal is affected 1) if the gauge slips from its original position and 2) if the patient changes position.
- 220.127.116.11 Calibration of strain gauges is difficult; consequently, the displayed data are more qualitative than quantitative.
- 18.104.22.168 The mercury in the Silastic tubing can develop bubbles that can affect signal integrity.
- 7.3.2 Piezoelectric belts are placed around the chest and/or abdomen. Limitations of piezoelectric belts include:
- 22.214.171.124 The quality and interpretability of the respiratory signal is affected if belts slip out of their original position.
- 126.96.36.199 Belts are generally not calibrated; consequently, the displayed data are more qualitative than quantitative.
- 7.3.3Inductance plethysmography is accomplished by placing elastic belts around the abdomen and chest of the patient. Limitations of inductance plethysmography include:(20)
- 188.8.131.52 The slipping of a belt from its original position affects the calibration, and, consequently, the quality and interpretability of the respiratory signal.
- 184.108.40.206 Calibration of inductance plethysmography is difficult in morbidly obese people, and, consequently, the data are more qualitative than quantitative.
- 220.127.116.11 The thin wires on the surface of inductance plethysmography belts are easily broken with a consequent loss of the monitored signal.
- 7.3.4 Impedance pneumography uses 2 or 3 electrodes attached to the patient in a configuration similar to that of a 3-lead ECG. Limitations of impedance pneumography include:(21)
- 18.104.22.168 The technique provides only a qualitative indication of chest-wall movement. There is no direct relationship to the volume of air within the chest.
- 22.214.171.124 Because impedance pneumography displays only global chest-wall movement, obstructive apneic events cannot be clearly differentiated from central or mixed apneic events in the absence of airflow measures.
- 126.96.36.199 No standards exist for the frequency response, precision, and reliability of signal reproductions that represent changes in thoracic impedance.
- 188.8.131.52 Signal degradation can occur with changes in body position.
- 184.108.40.206 Impedance pneumography is susceptible to cardiogenic artifact.
- 7.4 Record of intercostal muscle activity. Intercostal EMG is monitored by gluing or taping small metal-disk electrodes to the patient's intercostal spaces. When these electrodes are placed near the insertion of the diaphragm, they can monitor EMG activity from both the diaphragm and intercostal muscles. Limitations of intercostal EMG recording are the same as those listed in Section 7.1.
- 7.5 Record of esophageal pressure changes:
- 7.5.1 Esophageal pressures are monitored via esophageal balloons or liquid-filled catheters, which are attached to pressure transducers and signal amplifiers. Esophageal balloons are inflated with air or liquid, whereas liquid-filled catheters are filled with water or saline. They are inserted into the nares, through the nasopharynx, and into the esophagus to monitor changes in intrathoracic pressures occurring with inspiration and expiration.(22)
- 7.5.2 An esophageal balloon consists of a catheter attached to an air-filled balloon, which is inserted into the esophagus. Problems with esophageal balloons include:
- 220.127.116.11 inadvertent tracheal intubation;
- 18.104.22.168 nasal trauma or irritation;
- 22.214.171.124 inadequate filling rendering the pressure measurements inaccurate;
- 126.96.36.199 patient discomfort to a degree sufficient to alter sleep staging;
- 188.8.131.52 patient movement's altering the position of the catheter, and, thereby affecting the accuracy of the pressure measurements.
- 7.5.3 Liquid-filled catheters are similar to esophageal balloons. Limitations include:
- 184.108.40.206 air bubbles in the tubing can alter the characteristics of the pressure transmission through the tubing and cause the pressure readings to be inaccurate;
- 220.127.116.11 inadvertent tracheal intubation can give rise to irritation or aspiration of a catheter's contents into the lungs;
- 18.104.22.168 patient movement can alter the position of the catheter, thereby affecting the accuracy of the pressure measurements.
- 7.6 Record of airflow
- 7.6.1 Thermistors and thermocouples can provide information about the presence or absence of airflow based on temperature differences between inhaled and exhaled air. The devices incorporate small wiry sensors that can be taped below the patient's nostril and next to the mouth. Limitations include:(23)
- 22.214.171.124 the devices cannot be calibrated and, therefore, provide only qualitative information;
- 126.96.36.199 the collection of moisture on a sensor can affect its ability to detect temperature changes and affect signal integrity;
- 188.8.131.52 dislodgment from the subject results in partial or complete loss of the signal.
- 7.6.2 Pneumotachometers/pneumotachographs: Pneumotachography is used when quantitation of airflow and volume are necessary. Limitations of pneumotachography include:(24)
- 184.108.40.206 The patient's nose and mouth must be securely covered (leak free) by a face mask with the pneumotachometer attached to it;
- 220.127.116.11 The pneumotachometer and face mask are obtrusive and cumbersome and may not be tolerated by the patient and, therefore, may not be suitable for routine polysomnography;
- 18.104.22.168 Pneumotachography requires calibration at different air flows, and the amplifiers need to be calibrated frequently to ensure linearity.
- 7.6.3Recording of tracheal sounds audible via microphone is inexpensive and can be used to quantitate snoring.(7)
- 7.7Recording of oxygen saturation (SpO2). Pulse oximetry transmits two wavelengths of light through a pulsatile vascular bed to measure arterial oxygen saturation. Many factors can affect device accuracy and these have been well described.(25-29) With respect to polysomnography, limitations of pulse oximetry include:
- 7.7.1 During apneic events, the pulse oximeter may alarm and awaken the patient. Therefore, during polysomnography, it is important that the audible low-oxygen-saturation alarm be disabled. Alarms should be disabled only if a qualified technician is in constant attendance, monitoring the physiologic signals, and able to intervene if clinically indicated.
- 7.7.2 Not all pulse oximeters have an analog output that can be interfaced with the polygraph recorder.
- 7.7.3 Pulse oximetry does not reflect total gas exchange (cannot detect changes in PaCO2).
- 7.7.4 The polygraph amplifier requires calibration to ensure that the oximeter recording is linear and displayed within institutional requirements (ie, each drop in SpO2 causes a 1-mm pen deflection).
- 7.7.5 Lengthy signal averaging in some older pulse oximeters may produce imprecise values during acute events such as apnea.(29)
PSG 8.0 ASSESSMENT OF NEED:
In those patients who are suspected of sleep-related respiratory disturbances, periodic-limb-movement disorder, or other sleep disorders described within The International Classification of Sleep Disorders, Diagnostic and Coding Manual.(6) polysomnography is used to assess and quantify the presence and severity of such disturbances and their effect on oxygenation, cardiac status, and sleep continuity.
PSG 9.0 ASSESSMENT OF TEST QUALITY:
- 9.1 With respect to sleep-related respiratory disturbances, polysomnography should either confirm or eliminate a diagnosis.
- 9.2 Documentation of findings, suggested therapeutic intervention, and/or other clinical decisions resulting from polysomnography should be noted in the patient's chart.
- 9.3 Each laboratory should devise and implement indicators of quality assurance with respect to equipment calibration and maintenance, patient preparation and monitoring, scoring methodology, and intertechnician scoring variances.
PSG 10.0 RESOURCES:
- 10.1 Equipment:
- 10.1.1 A polygraph recorder capable of recording a minimum of 10 channels of high-frequency physiologic data. The data should be recorded on strip-chart paper, in ink (or by inkless pens writing on heat-sensitive paper) at a paper speeds from 10-30 mm/s. The polygraph may also be interfaced with an analog or digital storage device (cassette tape, optical disk) that has the ability to store and to print all raw data collected during the study.(30) The polygraph should be equipped with both alternating (AC) and direct-current (DC) bioamplifiers, with user-selectable electrical filters and sensitivities.
- 10.1.2 To record EEG and EOG, the polysomnographic recorder should have the sensitivity to give a pen deflection of 5.0-10.0 mm for a 50 microvolt signal. A calibration signal of 50 microvolt/cm is most common. It is essential that the recorder be calibrated prior to the study because changes in signal amplitude are one criterion for scoring the study.(13)
- 10.1.3 When recording EMGs, a pen deflection of 2 microvolt/mm is common; however, the absolute magnitude of the amplitude is irrelevant. Rather, the emphasis is placed upon relative changes in the EMG amplitude.(13)
- 10.1.4 EEG and EOG signals are amplified with an AC amplifier. The high-frequency-filter setting should be > or = 30-35 Hz with a low-frequency-filter setting of < or = 0.3 Hz.(13)
- 10.1.5 The EMG signal is also amplified with an AC amplifier. A high-frequency-filter setting of 70-120 Hz and a low-frequency-filter setting > or = 5 Hz are suggested.(13)
- 10.1.6 To record the rhythm ECGs, two additional EEG electrodes are configured for Leads I, II, III, or MCL. No particular emphasis need be placed on signal amplitude provided the signal is large enough to be discernible.
- 10.1.7 Output from respiratory effort and airflow-sensing devices and oxygen-saturation data are directed to the recorder and displayed with the other data.
- 10.1.8 Body position may be determined with mercury switches, with output signal directed to the recorder and displayed with the other data. If body position is monitored by direct observation, the patient's position should be noted at the beginning of the recording and whenever changes in position occur.
- 10.2 Personnel qualifications:
- Level-I personnel are designated as those with at least a high school education but without formal credentialing in polysomnographic technology and with no professional credentialing or licensing. Level-I personnel can be described as those who meet minimal job requirements and have received on-the-job polysomnographic training. Both Level-I and Level-II personnel should work under the direction of a physician specifically trained in the diagnosis and treatment of sleep disorders. Any personnel responsible for observing the patient should hold a current course-completion care in cardiopulmonary resuscitation (CPR) at, at least, the basic life support (BCLS) level and be competent to perform cardiac defibrillation.(31)
- 10.2.1 Level-I personnel:
- 10.2.1.1 may perform polysomnography;
- 10.2.1.2 need to be proficient in patient preparation, sensor application, and operation of the polysomnograph;(2)
- 10.2.1.3 should be familiar with operating principles and basic troubleshooting of the equipment listed in Section 7.0.
- 10.2.1.4 have an appreciation for the scoring and interpretation of polysomnograms and their use in the diagnosis of patients with sleep disorders;
- 10.2.1.5 have effective patient assessment and communication skills relative to recognizing and reporting adverse patient conditions (ie, a decline in the patient's clinical condition).
- 10.2.2 Level-II personnel
- 10.2.2.1 should have the knowledge and demonstrated ability to perform Level-I responsibilities,and
- 10.2.2.2 should be either credentialed or licensed as a Registered Polysomnographic Technician (RPSGT), a Registered Electroencephalographic Technologist (REEGT), Respiratory Care Practitioners (CRTT or RRT), or a Registered Nurse (RN);
- 10.2.2.3 must demonstrate the ability to score polysomnographic recordings and have an understanding of interpretative methods.
- 10.2.2.4 should have documented ability to recognize the patterns of polysomnographic variables as they are monitored and recorded enabling the practitioner to differentiate among technical difficulties and occult pathophysiology during polysomnography;
- 10.2.2.5 should display a proficiency in interpreting respiratory variables affected by initiation of continuous positive or bi-level positive airway pressure or nocturnal ventilatory support and should demonstrate knowledge of normative ranges for such variables as those measured during the monitoring of arterial blood gas, end-tidal CO2, oxygen saturation, respiratory movement, and airflow;(32)
- 10.2.2.6 should be able to initiate and titrate supportive therapy for sleep-related respiratory disorders;(32)
- 10.2.2.7 should be able to assess the patient's response to therapy;
- 10.2.2.8 should be able to recommend modifications to prescribed therapy, as appropriate;
- 10.2.2.9 should be familiar with operating principles, interface techniques, data acquisition, scoring, and reporting for other than cardiopulmonary-based studies, such as penile tumescence and seizure disorders.
PSG 11.0 MONITORING:
- 11.1 During polysomnography, the patient variables to be monitored are those listed in Section 2.2. Intervention is required if the physiologic signals are lost due to problems with instrumentation or become obscured by artifact.
- 11.2 Infrared or low-light videocameras and recording equipment should permit visualization of the patient, by the technician, throughout the procedure.(30)
- 11.3 Patients undergoing polysomnography are being evaluated for the presence of a chronic disease; therefore, their clinical status is unlikely to deteriorate acutely. However, the center-based polysomnographic studies described in this guideline require the presence of a technician throughout the study. Therefore, the technician should intervene if an acute change in physiologic status occurs and communicate these changes to appropriate medical personnel.
PSG 12.0 FREQUENCY:
- 12.1 A second polysomnographic study may be indicated
- 12.1.1 if the first study is technically inadequate due to equipment failure;
- 12.1.2 if the subject could not sleep or slept for an insufficient amount of time to allow a clinical diagnosis;
- 12.1.3 if initiation of therapy or confirmation of the efficacy of prescribed therapy is needed.
PSG 13.0 INFECTION CONTROL:
Practitioners should exercise Universal Precautions and precautions for prevention of the spread of tuberculosis as appropriate.(33,34)
Cardiopulmonary Diagnostics Focus Group
- 13.1 Nondisposable patient use items (eg, pneumotachometers, face masks, electrodes) should undergo cleaning and sterilization procedures as recommended by the manufacturer. If sterilization is not feasible, then high-level disinfection is warranted.
- 13.2 The syringe and flat-tipped needle used to inject transduction gel into the EEG, EOG, and EMG electrodes should be discarded after use.
- 13.3 With respect to body-position sensors, inductance and impedance pneumography, abdominal or thoracic strain gauges, or piezoelectric belts, no special precautions are generally required. Gas sterilization may be used if the sensors or belts become contaminated with body fluids.
- 13.4 Some thermistors are equipped with disposable sensors. If nondisposable sensors are used, they should be cleaned and subjected to high-level disinfection after use.
Kevin Shrake MA RRT, Chairman, Springfield IL
Pam Minkley RRT RPSGT(Special Consultant representing
the Association of Polysomnography Technologists), Lansing MI
Sue Blonshine RRT RPFT BS, Lansing MI
Robert A Brown BS RRT RPFT, Madison WI
Michael J Decker CRTT, Cleveland OH
Gregg L Ruppel MEd RRT, St Louis MO
Jack Wanger MBA RRT RPFT, Denver CO
- Indications and Standards for Cardiopulmonary Sleep Studies. American Thoracic Society consensus statement. Am Rev Respir Dis 1989; 139:559-568.
- Decker MJ, Smith B, Strohl KP. Center-based vs patient-based diagnosis and therapy of sleep-related respiratory disorders and the role of the respiratory care practitioner. Respir Care 1994 39(4);390-396.
- Smith JR. Transferring EEG polysomnography technology to the home environment. In: Miles LE, Broughton RJ (eds) Medical monitoring in the home and work environment. New York: Raven Press;1990:217-229.
- American Electroencephalographic Society. Guideline fifteen: guidelines for polygraphic assessment of sleep-related disorders (polysomnography). J Clin Neurophysiol 1994;11(1):116-124.
- Martin RJ, Block AJ, Cohn MA. Indications and standards for cardiopulmonary sleep studies. Sleep 1985;8 (4):371-379.
- American Sleep Disorders Association. The International Classification of Sleep Disorders, Diagnostic and Coding Manual. ASDA 1990.
- Parisi R, Santiago T. Respiration and respiratory function: techniques of recording and evaluation. In: Sleep disorders medicine: basic science, technical considerations, and clinical aspects (Chapter 9), S Chokroverty, editor. Boston: Butterworth-Heinemann 1994.
- ASDA Standards of Practice. Portable recording in the assessment of obstructive sleep apnea. Ferber R, Millman R, Coppola M, Fleetham J, Murray CF, Iber C, et al. Sleep 1994;17(4):378-392.
- Strohl KP, Cherniack NL, Gothe B. Physiologic basis of therapy for sleep apnea. Am Rev Respir Dis 1986; 134:791-802.
- Fletcher EC, Donner CF, Midgren B, Zielinski J, Levi-Valensi P, Braghiroli A, et al. Survival in COPD patients with a daytime PaO2 > 60 mm Hg with and without nocturnal hemoglobin desaturation. Chest 1992;101(3):649-655.
- Strohl KP, Chester CS. Polysomnography for breathing disorders in sleep. In: Nochomovitz ML, Cherniack NS (eds) Noninvasive respiratory monitoring. Churchill Livingstone Inc NY, 1986:153-166.
- Sullivan CE, Issa FG, Bruderer J, McCauley V, Bye PT, Grunstein R, Costas L. Treatment of cardiorespiratory disturbances during sleep. Interdiscipl Topic Geronto 1987;22:47-67.
- Carskadon MA, Rechtschaffen A. Monitoring and staging human sleep. In: MH Kryger, T Roth, WC Dement (eds) Principles and practice of sleep medicine. Philadelphia: WB Saunders, 1989:665-668.
- Kennan S. Polysomnographic techniques: an overview. In: Sleep disorders medicine: basic science, technical considerations, and clinical aspects (Chapter 6), S Chokroverty, editor. Boston: Butterworth-Heinemann 1994:82.
- Walczak T, Chokroverty S. Electroencephalography, electromyography, and electrocardiography: general principles and basic technology. In: Sleep disorders medicine: basic science, technical considerations, and clinical aspects (Chapter 7), S Chokroverty, editor. Boston: Butterworth-Heinemann 1994:97.
- Grass Model 8 Instruction Manual. Part II--theory and application. Grass Instrument Co, Quincy MA, October 1973.
- The Atlas Task Force. Recording and scoring leg movements. Sleep 1993, 16(8):748-759.
- Tyner FS, Knott JR, Mayer WB Jr. Chapter 10:131. In: Fundamentals of EEG technology, Vol 1: basic concepts and methods. New York: Raven Press.
- Kryger MH. Monitoring respiratory and cardiac function. In: MH Kryger, T Roth, WC Dement (eds) Principles and practice of sleep medicine. Philadelphia: WB Saunders 1994:984-1007.
- Ancoli-Israel S. Evaluating sleep with the portable modified Medilog/Respitrace system. In: Miles LE, Broughton RJ (eds) Medical monitoring in the home and work environment. New York: Raven Press 1990:275-294.
- Blunt JY. Impedance pneumography. In: Spacelabs Biophysical Measurement Series, Respiration. Redmond, Washington: Spacelabs Inc 1992:107-126.
- Simons JH, Giacomini A, Guilleminault C. Routine use of a water-filled catheter for measuring respiration during NPSG studies: an overview of the procedures and clinical utility. Sleep Research 22:1993.
- Arguin LJ, Hall EP. Signal processing and data display. In: Kacmarek RM, Hess D, Stoller JK (eds) Monitoring in respiratory care. St Louis: Mosby 1993:71-95.
- Myrabo K. Airway monitoring of adult, pediatric, and neonatal patients. In: Spacelabs Biophysical Measurement Series, Respiration. Redmond, Washington: Spacelabs Inc 1992:55-78.
- McCarthy K, Decker MJ, Stoller J, Strohl KP. Pulse oximetry. In: Kacmarek R., Hess D, Stoller JK (eds) Monitoring in respiratory care. St Louis: Mosby 1993:309-347.
- American Association for Respiratory Care. AARC Clinical Practice Guideline: pulse oximetry. Respir Care 1991;36:1406-1409.
- Brown DP, Cheung PW, Kenny MA, Shoene RB, Strohl KP, Decker MJ. Theoretical models and experimental studies of pulse oximetry. Proceedings of Ninth Annual Conference of the IEEE Engineering in Medicine and Biology Society, Boston MA, Nov 13-16, 1987.
- Cahan C. Decker MJ, Hoekje PL, Strohl KP. Agreement between noninvasive oximetric values for oxygen saturation. Chest 1990,97(4):814-819.
- Decker MJ, Strohl KP. Pulse oximetry. In: Spacelabs Biophysical Measurement Series, Respiration. Redmond, Washington: Spacelabs Inc 1992:79-93.
- American Sleep Disorders Association Accreditation Committee. Standards for accreditation of sleep disorders centers. Rochester, Minnesota: American Sleep Disorders Association 1994.
- American Association for Respiratory Care. AARC Clinical Practice Guideline: Defibrillation during resuscitation. Respir Care 1995;40(7):744-748.
- Sullivan CE, Grunstein RR. Continuous positive airway pressure in sleep disordered breathing. In: MH Kryger, T Roth, WC Dement (eds) Principles and practice of sleep medicine. Philadelphia: WB Saunders 1989:559-570.
- Centers for Disease Control. Update: Universal Precautions for prevention of transmission of human immunodeficiency virus, hepatitis B virus, and other bloodborne pathogens in health-care settings. MMWR 1988;37:377-382,387-388.
- Centers for Disease Control. Guidelines for preventing the transmission of tuberculosis in health-care settings, with special focus on HIV-related tissues. MMWR 1990;39(RR-17):1-29.
Agostoni E, Sant'ambrogio G, del Portillo Carrasco H. Electromyography of the diaphragm in man and transdiaphragmatic pressure. J Appl Physiol 1960;15(6):1093-1097.
American Sleep Disorders Association Standards of Practice Committee, Thorpy M, chairman. Practice parameters for the use of polysomnography in the evaluation of insomnia. Sleep 1995;18:55-57.
Berthon-Jones M, Sullivan CE. Ventilatory and arousal responses to hypoxia in sleeping humans. Am Rev Respir Dis 1982;125:632-639.
Gilliam PMS. Patterns of respiration in human beings at rest and during sleep. Bull Physiopathol Resp 1972;8: 1059-1070.
Haleász P, Ujszászi J, Gádoros J. Are microarousals preceded by electroencephalographic slow wave synchronization precursors of confusional awakenings? Sleep 1985; 8(3):231-238.
Laffont F, Autret A, Minz M, Beillevaire HP, Cathala HP, Castaigne P. Sleep respiratory arrhythmias in control subjects, narcoleptics and non-cataplectic hypersomniacs. Electroencephalogr Clin Neurophysiol 1978;44: 697-705.
Miles LE, Dement WC, Objective sleep parameters in elderly men and women. Sleep 1980;3(2):131-149.
Önal E, Lopata M, Ginzburg S, O'Connor TD. Diaphragmatic EMG and transdiagphragmatic pressure measurements with a single catheter. Am Rev Respir Dis 1981; 124:563-565.
Phillipson EA. Control of breathing during sleep (state of the art). Am Rev Respir Dis 1978;118:909-937.
Rechtschaffen A, Kales A (eds) A manual of standardized terminology: techniques and scoring systems for sleep stages of human subjects. Los Angeles: UCLA Brain Information Service/Brain Research Institute, 1968.
Sackner MA, Landa J, Forrest T, Greeneltch D. Periodic sleep apnea: chronic sleep deprivation related to intermittent upper airway obstruction and central nervous system disturbance. Chest 1975;67(2):164-171.
Sauerland EK, Orr WC, Hairston LE. EMG patterns of oropharyngeal muscles during respiration in wakefulness and sleep. Electromyogr Clin Neurophysiol 1981,21: 307-316.
Sauerland EK, Harper RM. The human tongue during sleep: electromyographic activity of the genioglossus muscle. Exp Neurol 1976;51:160-170.
Shepard JW Jr, editor. Atlas of sleep medicine. Rochester MN: Mayo Foundation 1991.
Interested persons may copy these Guidelines for noncommercial
purposes of scientific or educational advancement. Please credit AARC and Respiratory
Reprinted from the December 1995 issue of RESPIRATORY CARE [Respir Care 1995;40(12):1336–1343]